The DTQ450X from Component Design (CDN) is a quick read digital thermometer thermometer. It is NSF listed, waterproof and comes with a 5 year warranty.
Hands on Review CDN DTQ450X Quick Read Thermometer
In the packageBottom viewBattery DoorCompletely submerged. Note: The manual explicitly says not to do this. Guess what? I did it! I completely submerged this thermometer for version 1 of this review sometime around November of 2012. The thermometer has gotten plenty wet since then and is still performing like a champ. I’m not suggesting that you completely submerge yours. You should follow the manual’s instructions. I am saying, mine has held up to this type of treatment.
I’ve experienced different boiling temperature on different days using the same equipment and testing methods. Boiling temperature is a moving target. It is based on elevation and atmospheric conditions. The day of this test, boiling point was a consistent 211.8 deg F as read by my highly accurate Thermapen.The CDN DTQ450X reads 211.6 deg F. It’s reading to .2 deg F low. Well within it’s stated accuracy of +- 2 deg F. Practically speaking, for our needs, this is right on.Ice water is a more reliable reference point. Here is the CDN DTQ450X head to head with the Thermapen. As you can see, the Thermapen reads right at 32 deg F. The CDN DTQ450X reads 32.9 deg F. Again, well within it’s stated 2 deg F range. This is a good result. Not great, but good. That fact that it is more accurate at the higher range bodes well for homebrewers. We’re much closer to that end of the spectrum for most of the things we’re doing.
This video shows a response time test using the CDN DTQ450X. I tried to do this with water that was around mash temperature. I’m counting the start of the test as right when the metal basket hits the water. The probe of the thermometer is submerged at 2.28 seconds into the video. It levels out at 138.3 deg F at 5.7 seconds. That’s a response time of… 3.42 seconds. For .1 deg F resolution, that’s a great response time. In fact, it’s 1.98 seconds faster than a similar test I completed on my review of ThermoWorks Thermapen.
The DTQ450X measuring grain temperature
I’ve used the CDN DTQ450X since October of 2012. This same unit continues to look and function well. It’s also a key component of my Brew Day Box. It displays very good accuracy and it has a very, very good response time. If you’re looking for a thermometer for homebrewing and general use this is an excellent choice that I can recommend based on experience.
What are Other’s Saying? Search this product’s Amazon reviews for “brew” – may include reviews for other sizes or variations
CDN DTQ450X ProAccurate Quick-Read Thermometer – via Amazon
More Homebrew Finds!
- Last 50 Finds!
- Top Deals – a curated list of the best deals
- Homebrew Reviews – one of the largest libraries of homebrew reviews in existence!
- Our Top Posts – tips, how-tos, resources posts and more
- Let’s be Friends!
Make sure the components you use are compatible and rated for your intended application. Contact manufacturer with questions about suitability or a specific application. Always read and follow manufacturer directions. review:cdndtq450x tag:tpr rs:5