Here’s a hands on review of ThermoWork Classic Thermapen Thermometer including response time and accuracy tests.
Unfolded. This automatically turns on and off as the probe is unfolded and folded back in. I was a bit surprised by how large this is. I think that’s a good thing. The larger body size makes it easy to hold on to. It also feels solid and well made.
The Thermapen measuring boiling water. It reads 211.8. The boiling point of water varies based on elevation/barometric pressure and mineral content of the water. On two separate days, I’ve gotten two consistent and different readings for the boiling point of water. The last time I tested, it was consistently 211.4 deg F. The day this picture was taken with the water I was using, the boiling point of water was consistently 211.8 deg F. Although I’m sure you could figure out what the boiling point of water should be at any one point in time based on atmospheric conditions, it’s really… a moving target. Ice water is a much more consistent temperature. The good news for calibrating thermometers at boiling temperature is that, for homebrewers, a variance of .4 to .6 isn’t usually a big deal.
How do you know that your highly accurate Thermapen (+- .7 deg F) with NIST Traceable Calibration is really accurate? You use a even more accurate Reference Thermapen (+- .07 deg F) that is calibrated to 5 points and reads to the nearest 1/100th of a degree. This shows the Thermapen next to the Reference Thermapen, reading the same temperature. By the way… I wouldn’t recommend using a Reference Thermapen on brew day. What you gain in accuracy, you give up in response time. It’s about 20 seconds, which is great for 5 point accuracy, but an unneeded trade off for most brewing applications. The Reference model is good for double checking and calibrating other thermometers.
Test 1 – This video shows a response time test using the Thermapen. I tried to do this with water that was around mash temperature. I’m counting the start of the test as right when the metal basket hits the water. The probe of the Thermapen is submerged at 3.42 seconds into the video. It levels out at 149 deg F at 4.98 seconds. That’s a response time of… 1.56 seconds. That’s crazy. It’s like this thing knows what you’re going to put it into and uses xray vision to start testing as it’s traveling.
Test 2 – By default, the Thermapen displays whole degrees. I changed that setting to tenths and did another response time test. The Thermapen is submerged at 1.06 seconds. It levels out at 130.4 deg F at 6.46 seconds. That’s a response time of 5.4 seconds. It’s not quite as eerie as the whole degree test, but it’s still an excellent response time.
Here is the Thermapen checking the temperature on the grain bill for More Beer’s Citra Pale Ale Kit.
- Range -58.0 to 572.0°F (-49.9 to 299.9°C)
- Accuracy ±0.7°F (±0.4°C) from -58 to 392°F (-49.9 to 199.9°C), ±1.8°F (±1.0°C) from 392 to 572°F (200 to 299.9°C)
- Resolution 0.1°F/°C full range (user reconfigurable to 1.0°F/°C)
- Response Time From 75°F to 32°F: ≤ 3 seconds to within 1°F
- Probe 4.5 L inches, High Performance Type K Thermocouple
- Display 0.55 inches (14mm) LCD
- Auto Shutoff 10 Minutes (user reconfigurable to disable auto-off)
- Environmental Range -4 to 122°F (-20 to 50°C)
- Battery CR2032 (3V) lithium coin cell x 2, 1500 hours
- Dimensions 1.85 H x 6.1 W x 0.75 D inches (47 H x 153 W x 19 D mm)
- Weight 0.25 lbs (97g)
- Certificate Includes NIST-Traceable calibration certificate
Because of my work on Homebrew Finds, I own a lot of thermometers. A lot. Overall, this is one of the nicest brewing thermometers I own. It’s extremely feels very well built, it has an amazing response time and it is spot on accurate. This is a great brewing thermometer.
I’ve used this for years and can wholeheartedly recommend it to you.
Recent Great Deals [view more]:
- Bell’s General Store – Popular Hops Sale – 1 lb Cascade –…
- Reader Tip: 4 lb PBW – $23.25
- 25′ Stainless Steel Immersion Chiller – $37.06 + $5.93 shipping
- The OxyWand Oxygenation Kit – 0.5 Micron – $53.99 + Free Shipping